Disconnect the NIBP hose from the connector on the front chassis. Message: Out of paper. Error Code: C22 - Temperature time limit exceeded. Touch Clear to delete all data. http://txtbl.com/welch-allyn/welch-allyn-error-codes-c05.html
Causes: Patient measurements have changed enough to require attention. Temperature cable connector slot Temperature cable connector d. Radio out of network range. Page 20 Functional Veriﬁcation Procedure Functional Veriﬁcation Monitor-Off Current With the monitor powered down, verify that the current draw from the power supply is less than 1 mA. this website
Causes: An unqualified mattress type is in use Action: Switch to a qualified mattress type Message: Cannot measure respiration. The valve opens to release pressure. 7. If the problem persists, call for service. Action: Call for service to calibrate the NIBP module.
Message: Unable to detect new temperature. Remove the temperature-board assembly from the SureTemp Plus module as follows: a. Oldest record overwritten. Welch Allyn 52000 Service Manual Action: Replace the paper.
Call for service. Incompatible SpO2sensor.Improperlyconnected SpO2sensor.Faulty SpO2sensor.Faulty SpO2board.Use a compatible sensor.Properly connect the sensor cable.Replace the sensor.Replace the SpO2board.Repeated C20 temperatureerror code.Probe well is not installed in theprobe-well housing.Faulty probe.Faulty temperature board.Install the probe Message: CO2 temperature out of range. hop over to this website If the error persists, return themonitor to an authorized service center forreplacement of the main board.Error code E40 NIBP module failure.
Page 58: Disassemble And Remove The Nibp Assembly Disassemble and Remove the NIBP Assembly Repair or replacement of the NIBP assembly is not supported. Welch Allyn 53n00 Message: Incorrect NIBP cuff size; check patient type. Action: Adjust the user settings and retry. Action: Remove the sensor from the patient and reapply.
If problem persists, replace probe. http://personalcare.manualsonline.com/manuals/mfg/welch_allyn_medical_diagnostic_equipment/300_series.html?p=38 Device will power down. Welch Allyn 300 Series Service Manual Radio Messages: Message: Radio not functional. Welch Allyn 53nto If problem persists, replace probe.
Action: Stop continuous measurements. http://txtbl.com/welch-allyn/welch-allyn-52000-error-codes.html Error Code: C9 - SpO2 time limit exceeded. Verify all cable connections, and then cycle thepower. Check tubing for kinks and connection integrity. Welch Allyn 420 Service Manual
Action: Connect a sampling line to the monitor. If problem persists, contact Technical Service. Causes: The clinician ID does not match an ID in the network. this content Causes: The patient ID does not match an ID in the patient list or network.
Seller's Information: Portland Medical LLC view seller's other items: viewall Need to ship this item? Welch Allyn 53nto Vital Signs Monitor Action: Add a patient ID. Page 38 Troubleshooting Chart Troubleshooting and Repair Symptom Possible Cause Suggested Action Error code E31 RAM test failure.
Verify SpO2 and pulse rate using an alternative method. Page 63 Disassembly Procedure Remove and Disassemble the Printer Assembly Push the printer legs very slightly outward while pulling them toward you. Action: Resume using the monitor. Welch Allyn Vital Signs Monitor 6000 Series Front Panel Controls Set alarm limits Power on/off Silence alarms Print patient data Scroll up/down Review patient data Scroll forward/back Increase/decrease value (The scroll icon appears as these two arrows in
Blood Pressure: Error Code: C02 - Auto-zero failure. Hide thumbs 1 2 3 4 5 6 7 8 9 10 11 12 13 14 15 16 17 18 19 20 21 22 23 24 25 26 27 Error Code: bv - Internal temperature malfunction. http://txtbl.com/welch-allyn/welch-allyn-error-codes.html Unless otherwise noted, the assembly procedure is simply the reverse of the disassembly procedure.
Message: Stop patient intervals to select new patient. Error Code: E0.1 - Probe heater error. C02 might not be accurate. Change the cuff inflation target (CIT).
Call for service. b. Error Code: C22 - 10-minute diagnostic time exceeded. Change ambient temperature.
Service Manual Vital Signs Monitor Series 300... LOT PURCHASE PRICE is only $500. Printer Reassembly Notes a. Message: Printer door is open; close to continue.