Sign up! View Cart (0) Order Status Lookup RMA Status Lookup Warranty Lookup Find a Distributor Contact Sales Contact Welch Allyn Contact via Phone Products ‹ Back Physical Exam Physical Exam Sets and Learn More Discontinued Products Physical Exam Physical Exam Sets and Systems Eye Exam Ear Exam Nose and Throat Exam Stethoscopes Endoscopic Exam Womens Health Power Handles Patient Monitoring Vital Signs Monitor View Cart (0) Distributor Tools Students Careers About News Your cart is empty. check over here
If the problem persists contact Welch Allyn Customer Care. All rights reserved. | | North America United States | English Canada | English Latin America | SpanishAsia & Pacific Rim JAPAC | English Australia | English New Zealand | English Try another cycle. Print This PagePrint ShareShare Url of this page: HTML Link: Bookmark Manuals Brands Welch Allyn Manuals Thermometer SureTemp Plus REF 690 Service manual Welch Allyn SureTemp Plus 690 Service Manual: Error
Generated Tue, 01 Nov 2016 14:26:10 GMT by s_sg2 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.10/ Connection Learn More Stress-Test Systems Holter Monitoring Systems Ambulatory Blood Pressure Monitor View All Digital Blood Pressure Devices > Featured Product: ABPM 7100 Ambulatory Blood Pressure Monitor An easy-to-use 24-hour ambulatory blood Sure Temp® Plus Service Manual × Show quick links Table of Contents ErrorCodes:Troubleshooting Previous page Next page 1...202122232425262728 Also See for Welch Allyn SureTemp Plus 690 Welch Allyn SureTemp Plus If the problem persists contact Welch Allyn Customer Care.
Make sure the tip temperature is below 110ºF/43.3ºC. Products All Physical Exam Patient Monitoring Blood Pressure Measurement Thermometry Cardiopulmonary Medical Lighting Animal Health Population Health Management Technologies Factory Refurbished Promotions Welch Allyn Home Parts & Accessories Batteries Lamps Cables Try another predict cycle. Welch Allyn Suretemp Plus Anti Theft Your new...
I can't find any information on how to reset this error message after I check the calibration.Does anyone know how to do the CAL or reset it. Welch Allyn Suretemp Plus 690 Generated Tue, 01 Nov 2016 14:26:10 GMT by s_sg2 (squid/3.5.20) ERROR The requested URL could not be retrieved The following error was encountered while trying to retrieve the URL: http://0.0.0.9/ Connection If the problem persists replace the probe or contact Welch Allyn Customer Care. https://www.manualslib.com/manual/846757/Welch-Allyn-Suretemp-Plus-690.html?page=24 E4.4 Device EEPROM starting handshake failure.
If the problem persists replace the probe. http://mobile.medwrench.com/thread/51778/error-code Learn More Spirometry Cardiopulmonary Data Management Medical Lighting Minor-Procedure Lighting Exam Lights View All Exam Lights > Featured Product: Green Series 300 General Exam Light A single white, bright LED that Welch Allyn Suretemp Plus Troubleshooting Guide If the problem persists contact Welch Allyn Customer Care. Welch Allyn Suretemp Plus 692 Verify that the batteries are fresh.
Verify that the batteries are fresh. check my blog Learn More Sphygmomanometers DuraShock Platinum Series Sphygmomanometers DuraShock Gold Series Sphygmomanometers DuraShock Silver Series Sphygmomanometers DuraShock Bronze Series Sphygmomanometers 767 Wall and Mobile Aneroids Tycos Aneroids DuraShock Hand Aneroids Home Blood E4.7 Can't initialize the device EEPROM E4.8 Device was not calibrated E5.0 Heater is not working properly. 20 ® Customer Care at 1-800-535-6663. Your cache administrator is webmaster. Welch Allyn Thermometer Anti Theft System
E4.6 Transmit failure to the Device EEPROM. Learn More Retinoscopes Autorefractor Ophthalmoscopes, Binocular Indirect Ear Exam Otoscopes, Traditional Otoscopes, Pneumatic Otoscopes, Operating Otoscopes, Pocket Tympanometric Systems Audiometers Otoscopes, Macroview OAE Hearing Screener Ear Wash System Nose and Throat Try another cycle. this content North America United States | English Canada | English Latin America | SpanishAsia & Pacific Rim JAPAC | English Australia | English New Zealand | English China | Chinese Japan |
Learn More Headlights Animal Health Ear Exam Otoscopes, Macroview Otoscopes, Traditional Otoscopes, Pneumatic Otoscopes, Operating Eye Exam Ophthalmoscopes, Wide-View Direct Ophthalmoscopes, Traditional Direct Ophthalmoscopes, Binocular Indirect Nose and Throat Exam Illuminators Welch Allyn Suretemp Plus How To Change Battery Edited by Huw (15/01/12 09:19 PM)Edit Reason: Created new topic for this question. Learn More Thermometry Oral/Axillary/Rectal Thermometers Ear Thermometers View All Ear Thermometers > Featured Product: Braun ThermoScan PRO 6000 The New Welch Allyn Braun ThermoScan PRO 6000 e...
GET STARTED © 2015 Welch Allyn®. Troubleshootingpage 22................................................................................................................................................................ View Cart (0) Order Status Lookup RMA Status Lookup Warranty Lookup Find a Distributor Contact Sales Contact Welch Allyn Contact via Phone Products ‹ Back Physical Exam Physical Exam Sets and Sure Temp Liquid Thermometer Inside there is maybe a chip that needs resetted as they have a tendancy to subjected to the drop test on the ward.The guys at welch allyn are good and will
I have the service manual but it doesn't contain schematics as earlier models did.We don't have a Welch Allyn office in NZ only a representing company thay sells their products but The other thing you may try if you havent is to change the probe and restart it. E4.2 Reference power supply malfunction. have a peek at these guys Answer: Please refer to your SureTemp® Plus 690 or SureTemp® Plus 692 thermometer thermometer User Manual for an explanation of error codes.
Verify the batteries are fresh. Your cache administrator is webmaster. The data is streaming out of the EEPROM before this point.Keith Top #59826 - 23/01/12 01:06 AM Re: Welch Allyn SureTemp PLUS 692 Thermometers [Re: Mack] Mack Novice Registered: 21/09/10 Posts: If the problem persists replace the part that isindicated by the flashing icon Answer provided by Welch Allyn.Bob4 years agoReply from Bob This is a probe error.
E4.3 Device EEPROM save failure. try a new Probe.«1» Welch Allyn - Sure Temp Plus 690by Welch AllynProduct DetailsDESCRIPTIONEasy to use with intuitive operation, reduces the risk of cross-contamination with removable probe well and provides more E4.0 PTB measurement out of range. The system returned: (22) Invalid argument The remote host or network may be down.
Answer: A 5.0 error on your SureTemp Plus 690 or SureTemp Plus 692 thermometer means the heater is not working properly.