Causes: A hardware failure occurred in the power supply. If problem persists, contact Technical Service. If problem persists, contact Technical Service. Action: Check inflation settings and change as necessary. check over here
Learn more You're viewing YouTube in English (United Kingdom). Even when plugged in these will not work properly if the battery is going bad. Check patient condition and sensor position/contact. Action: Replace the cable. try this
Action: Turn the sensor right-side up. Action: Call for service. Device will power down.
Message: Check calibration gas concentration. Sign up! If problem persists, replace probe. Welch Allyn Spot Vital Signs Monitor Message: Unable to establish network communications.
Loading... Welch Allyn Spot Vital Signs Lxi Set Date And Time Action: Check the cuff, tubing, and connections for leaks. Causes: The SureTemp temperature module is faulty. Message: Check for the occlusion in gas line.
Call for service.
Message: Incorrect NIBP cuff size; check patient type. Welch Allyn Error Codes Action: Resume using the monitor. Welch Allyn Spot Vital Signs 420 Series Date and time not set.
Limit patient motion. http://txtbl.com/welch-allyn/welch-allyn-vital-signs-monitor-error-codes.html Message: Calibration failed. Power down and restart. Working... Welch Allyn Spot Vital Signs Battery
Action: Connect a sampling line to the monitor. Welch Allyn Error Code E43 Connex Error Message List: Patient movement messages: Message: Replace the bed sensor. Replace the bed sensor.
If problem persists, contact Technical Service. Error Code: bv - Internal temperature malfunction. Language: English (UK) Content location: United Kingdom Restricted Mode: Off History Help Loading... have a peek at these guys Error Code: P - Loss of tissue contact.
Error Code: P - The temperature probe has poor tissue contact. To use the AC power adapter: 1. Lost or unstable signal. Ethernet Messages: Message: Network not found; check network cable connection.
Message: Unable to identify patient. Check patient condition. If the problem persists, replace the probe.