C6 SpO2 pulse rate out of range Check your patient’s condition. Causes: The configuration required patient ID to save data. Action: Check the patient type and cuff size. Verify patient type setting. check over here
Contact Technical Service. To print the manual completely, please, download it. Air leak. Message: Tube type does not match device configuration. (NIBP measurement is available) Causes: The tube connected to the NIBP sensor does not match the monitor's configuration.
Action: Reattach the cable to the monitor. Contact Technical Service. it means it is unable to take blood pressure.
Action: Touch OK to dismiss or connect the monitor to AC power. Causes: Battery power is low. Unable to determine blood pressure. Welch Allyn E33 Error Code Message: Replace temperature probe.
Causes: The sensor is turned 90 degrees, place veritcally (Low confidence condition) Action: Place the sensor horizontally under the patient's mattress with the cable extending toward the head of the bed. Welch Allyn Spot Vital Signs Battery Action: Call for service. Temperature: Error Code: C20 - Broken/missing probe. Message: Stop patient intervals to select new patient.
Causes: Excessive ambient or environmental noise. SpO2 Error Codes E7 Internal error. Welch Allyn C13 Error Message: Out of paper. Welch Allyn Error Code E43 Error Code: E11 - Cuff pressure duration exceeded.
Message: Retry temperature measurement. (Note: This message often accompanies other temperature messages.) Causes: A probe heater or data error occurred. http://txtbl.com/welch-allyn/welch-allyn-52000-error-codes.html Causes: An unrecognizable device is connected. Radio Messages: Message: Radio not functional. View Cart (0) Distributor Tools Students Careers About News Your cart is empty. Welch Allyn Spot Vital Signs 420 Series
Messages: Unable to identify clinician. Error Code: C05 - Excessive noise or air leak. E20 - E50 General internal malfunction Call Technical Service. http://txtbl.com/welch-allyn/welch-allyn-error-codes.html Description Blood pressure reading cancelled by user.
Message: SpO2 sensor expires in … Causes: The SpO2 sensor will expire soon. Welch Allyn Blood Pressure Monitor Error Codes Error Code: C01 - Blood pressure reading cancelled by user. North America United States | English Canada | English Latin America | SpanishAsia & Pacific Rim JAPAC | English Australia | English New Zealand | English China | Chinese Japan |
Causes: An occlusion was detected in the sampling line, which automatically initiated a purge cycle. What Our Users Say Press & Media Contacts Advertising DMCA Policy Brands × Login Login to ManualsLib Don't have an account? Check sensor. Welch Allyn 420 Series Battery Retake blood pressure reading.
Action: Check the bed sensor cable to ensure that connections to the monitor and the extension cable are tight. Learn More Thermometry Oral/Axillary/Rectal Thermometers Ear Thermometers View All Ear Thermometers > Featured Product: Braun ThermoScan PRO 6000 The New Welch Allyn Braun ThermoScan PRO 6000 e... Replace probe. http://txtbl.com/welch-allyn/welch-allyn-vsm-300-error-codes.html Error Code: C8 - Faulty SpO2 sensor.
Use wall transformer. Limitpatient motion. Causes: The monitor is set to take interval readings. Check for hose constriction.
Error Code: P - The temperature probe has poor tissue contact. Change ambient temperature. Contact Technical Service. Loading...
Device will power down. what is error 38 on spot vital welch allyn?