Causes: The date or time is not set properly - Action: Reset the date or time. Action: Call for service. Action: Replace the sensor. Action: Replace the sensor. check over here
If problem persists, contact Technical Service. Causes: User settings require adjustment. She is interested in ordering these for our surgical floor. Table 10. http://www.welchallyn.com/en/service-support/answer-center/troubleshooting/cshm120-spot-shows-e42-error.html
Low battery level. Description Blood pressure reading cancelled by user. If problem persists, call for service. Action: Check the error message and take corrective action presented.
Action: Remove the sensor from the patient and reapply. Replace the SpO2 sensor. Learn More Headlights Animal Health Ear Exam Otoscopes, Macroview Otoscopes, Traditional Otoscopes, Pneumatic Otoscopes, Operating Eye Exam Ophthalmoscopes, Wide-View Direct Ophthalmoscopes, Traditional Direct Ophthalmoscopes, Binocular Indirect Nose and Throat Exam Illuminators Welch Allyn Spot Vital Signs Battery Action: Replace the temperature probe.
If problem persists, contact Technical Service. Welch Allyn Error Codes Causes: The sensor is turned 90 degrees, place veritcally (Low confidence condition) Action: Place the sensor horizontally under the patient's mattress with the cable extending toward the head of the bed. Message: Unable to establish network communications. http://www.fixya.com/support/t21070792-m_getting_error_e42_in_welch_allyn_spot Blood Pressure Cuff Pops OffPossible CauseExplanation and Corrective Action Inappropriate blood pressureDetermine blood pressure cuff size with the blood pressure cuff markings or seecuff size“Chart for Determining Blood Pressure Cuff Size” on
Video kiralandığında oy verilebilir. Welch Allyn Spot Vital Signs Lxi Service Manual Loading...Oops :Please try again. Error Code: E42 - Internal communications error. GET STARTED © 2015 Welch Allyn®.
Causes: The sensor is not under the patient's chest (low confidence condition) Action: Reposition the sensor under the patient's chest. Press the Mode button once to save the change and press the Power button to exit the Internal Configuration Mode. Welch Allyn C13 Error Replace probe. Welch Allyn Spot Vital Signs 420 Series C6 SpO2 pulse rate out of range Check your patient’s condition.
Causes: The SureTemp temperature module is faulty. check my blog Tim Root 366.883 görüntüleme 28:35 The Welch Allyn Connex® Spot Monitor Training - Süre: 7:06. After reconnecting the battery, reset the date and time. Learn More Discontinued Products Physical Exam Physical Exam Sets and Systems Eye Exam Ear Exam Nose and Throat Exam Stethoscopes Endoscopic Exam Womens Health Power Handles Patient Monitoring Vital Signs Monitor Welch Allyn Error Code E43
C04 Too long to inflate Check your BP hose and cuff for leaks. Device will power down. You may NOT copy or distribute the content that appears on this site without written permission from Fixya Ltd. © 2005-2015, Fixya, Ltd. http://txtbl.com/welch-allyn/welch-allyn-e38-error-code.html Use the Next Patient/Clear/Cancel or Blood Pressure Start/Stop buttons to change the Date/Time setting. 5.
USB and USB flash drive Messages: Message: USB Communication failure. Welch Allyn C05 Error Action: Save the measurements. Action: Replace the patient cable or sensor.
Action: Call for service to modify advanced settings. Error Code: E0.4 - Probe is over temperature. Contact Technical Service. Welch Allyn Spot Vital Signs Monitor Call for service.
Your new... Message: Retry temperature measurement. (Note: This message often accompanies other temperature messages.) Causes: A probe heater or data error occurred. Contact Technical Service. have a peek at these guys Causes: The bed sensor has expired.
General Error Codes Code C12 C13 E30 E31 E32 E33 E38 E42 E44 E45 Table 11. Causes: The sensor is not under the patient's chest (low confidence condition) Action: Adjust the sensor under the patient's chest. Generated Tue, 01 Nov 2016 09:21:09 GMT by s_hp90 (squid/3.5.20) Error Code: E0.1 - Probe heater error.
Causes: You or some other user interrupted the calibration. Replace probe. Oturum aç 12 0 Bu videoyu beğenmediniz mi? Causes: Missing, improperly inserted, or incompatible USB flash drive.
Check blood pressure cuff and tubing connections. Action: Connect the monitor to AC power. Disconnect the battery an dwait 5 minutes.